A simple, selective, rapid, precise and economical reverse phase HPLC method has been developed for the determination of melphalan in tablet. The analyte was resolved by using a mobile phase (Acetonitrile, water and 1% ortho phosphoric acid in the ratio 70:27:3 v/v/v) at a flow rate 1 ml/min on an isocratic HPLC system (PEAK) consisting UV-Visible detector, ODS C-18, RP column (4.6 mm i.d x250 mm) at a wavelength of 275 nm. The linear dynamic range for melphalan was 2.0 μg/mL – 14.0μg/mL. The limit of detection [LOD] and Limit of quantification [LOQ] for melphalan was 0.5μg/mL and 1.5μg/mL respectively.
Comments: 5 Pages. OK
[v1] 2019-05-19 06:02:31
Unique-IP document downloads: 15 times
Vixra.org is a pre-print repository rather than a journal. Articles hosted may not yet have been verified by peer-review and should be treated as preliminary. In particular, anything that appears to include financial or legal advice or proposed medical treatments should be treated with due caution. Vixra.org will not be responsible for any consequences of actions that result from any form of use of any documents on this website.
Add your own feedback and questions here:
You are equally welcome to be positive or negative about any paper but please be polite. If you are being critical you must mention at least one specific error, otherwise your comment will be deleted as unhelpful.