The new method of the high-fidelity measuring of resistances was proposed. A Wheatstone bridge was used with two unknown base resistors of R1 and R2 in one branch, the resistance decade boxes and unknown resistance Rx was plugged in other branch. By the selection of Rm1 of resistance decade boxes bridge was balanced. R1 and R2 switched places then. By the selection of Rm2 of resistance decade boxes the bridge was balanced in second times. Unknown resistance settled accounts on the formula of Rx=(Rm1•Rm2)1/2.
Comments: 5 Pages. 5 р. Russian. Odessa State Enveronmental University
[v1] 2017-10-22 15:27:42
Unique-IP document downloads: 15 times
Vixra.org is a pre-print repository rather than a journal. Articles hosted may not yet have been verified by peer-review and should be treated as preliminary. In particular, anything that appears to include financial or legal advice or proposed medical treatments should be treated with due caution. Vixra.org will not be responsible for any consequences of actions that result from any form of use of any documents on this website.
Add your own feedback and questions here:
You are equally welcome to be positive or negative about any paper but please be polite. If you are being critical you must mention at least one specific error, otherwise your comment will be deleted as unhelpful.