Authors: Kenneth C. Johnson
A compact and inexpensive EUV source using high harmonic generation can provide illumination for a tabletop EUV microscope, but such microscopes typically use only one of the source’s available harmonic wavelengths. This paper outlines a microscopy system that focuses multiple harmonic wavelengths onto distinct, spatially-separated focal spots on an inspection sample, which is scanned to generate multi-spectral, diffraction-limited EUV images of the sample. The system uses diffraction optics to separate the EUV harmonics and also to separate the source’s IR wavelength out of the EUV illumination.
Comments: 13 Pages.
[v1] 2015-12-22 02:31:47
Unique-IP document downloads: 101 times
Vixra.org is a pre-print repository rather than a journal. Articles hosted may not yet have been verified by peer-review and should be treated as preliminary. In particular, anything that appears to include financial or legal advice or proposed medical treatments should be treated with due caution. Vixra.org will not be responsible for any consequences of actions that result from any form of use of any documents on this website.
Add your own feedback and questions here:
You are equally welcome to be positive or negative about any paper but please be polite. If you are being critical you must mention at least one specific error, otherwise your comment will be deleted as unhelpful.